Fluctuating Redox (ORP) readings
A fluctuation in Redox (ORP) readings is not necessarily a problem. Given that it is an indicator of the rate of disinfection in your pool, it will almost never be exactly the same in between each measurement (ICO takes a reading every hour). However, if you notice large fluctuations (a change > 200mV between readings) that does not correspond to any treatment carried out, this could reveal a problem with your ICO or your pool.
|If you notice large, irregular fluctuations, this could be down to a contact problem between the sensor and ICO’s circuit board.
To resolve this : Widen the contact strips on your ORP sensor (Orange/Yellow)
|If the fluctuations are more regular, this could be due to an issue with the settings of your pool equipment (filtration system, electrolyser, etc…) Click here for the solutions.|
Was this page helpful?
[Total: 0 Average: 0]